MID Directive 2004/22/EC
LE-01M under the influence of current flow and applied voltage makes precise measurement of the amount of consumed electricity. Energy consumption is indicated by a flashing LED (1600 pulse / kWh) and its value is determined by the LCD display. Decimals represent hundredths of kWh (0.01 kWh = 10 Wh).
Communication with the rate of working as slave devices is carried out according to the standard Modbus RTU via RS-485. Read register values after conversion kWh give a result consistent with the indications on the display indicator. Each of the indicators is identified by a unique address transmitted by the user.
The indicator has a pulse output SO+ - SO-. This allows you to connect another device pulse-reading (SO) pulses generated by the counter. For proper operation of the meter is not required to connect additional devices.
The program allows to test reading of the calculated energy and make the settings of the basic parameters of the counter.
PC-LE communication via USB converter WE-1800-BTâ–ş
Programs for remote reading of electricity meters
LE-01M, LE-01MP, LE-01MQ, LE-03M, LE-03MP, LE-03M CT
The MeternetPRO application allows remote reading of states and indications of counters, multimeters, measuring transducers, expansion modules of input/output and other measuring devices communicating through Modbus RTU protocol. The exchange of data between devices and the application is executed through RS-485 or local area network (LAN). The application, along with the database is installed on a special MT-CPU-1 server, which operates in a local network. Software user interface is a web application (web site). Access to the application is executed through any web browser. In a LAN with a public IP address, you can configure the application and read the data over the Internet.
Subscription billing of electricity consumption module
Module of the subscription billing of the power consumption or other recorded incremental value, for example: water, heat, etc. It allows you to calculate the increases in the value in the designated billing periods (intervals). Cycles: monthly, weekly, daily, hourly. The module allows you to create multiple individual and parallel reports.
Program allows remote reading via RS-485 network display up to 1000 indicators.
Exchange of data between the counters and the application is done via a standard RS485 <-> USB port or server on the LAN (Ethernet). For LAN connection to the router with a static IP address, you can read the data via the Internet.
|Communication protocol||MODBUS RTU|
|Compatibility||MID Directive 2014/32 / EU|
|Reference voltage||230 V AC Â±30%|
|Current current||10:00 AM|
|Maximum current||100 A|
|The minimum measured current||0,04 A|
|Counter's own consumption||10 VA; 2 W|
|Scope of indication of the abacus||0Ă·99999,99 kWh|
|Counter constant||(0,625 Wh/imp) 1600 imp/kWh|
|Reading signaling||LED red|
|Pulse output SO + SO-||open collector|
|Connection voltage SO + SO-||max. 27 V DC|
|Connection current SO + SO-||max. 27 mA|
|Pulse time SO + SO-||34Ă·80 ms|
|terminal||screw terminals 25 mmÂ˛|
|dimensions||4,5 moduleu (75 mm)|
|Assembly||on rail 35 mm|
|Level of security||IP20|